Biosystem Bts 310 User Manual
DOWNLOAD >> https://blltly.com/2tzNui
How to Use and Maintain the BioSystems BTS 310 Analyzer
The BioSystems BTS 310 Analyzer is a spectrophotometer that can perform various biochemical and enzymatic tests on blood samples. It is designed for clinical laboratories and research centers that need fast and accurate results. The analyzer has a user-friendly interface, a large LCD screen, and a built-in printer. It can store up to 1000 test results and 100 calibration curves in its memory.
In this article, we will explain how to use and maintain the BioSystems BTS 310 Analyzer, based on the official user manual[^1^] and service manual[^2^]. We will also provide some troubleshooting tips and contact information for technical support.
How to Use the BioSystems BTS 310 Analyzer
To use the BioSystems BTS 310 Analyzer, you need to follow these steps:
Turn on the analyzer by pressing the power switch on the back panel. Wait for the self-test to complete.
Select the desired language from the menu by using the arrow keys and pressing ENTER.
Load the reagents and samples into the cuvette holder. Make sure they are properly labeled and aligned.
Select the test you want to perform from the menu by using the arrow keys and pressing ENTER. You can also create your own test by entering the parameters manually.
Enter the sample ID and other information by using the numeric keypad and pressing ENTER.
Press START to begin the analysis. The analyzer will automatically measure the absorbance of the sample at different wavelengths and calculate the result.
Wait for the result to appear on the screen. You can also print it by pressing PRINT.
Remove the cuvette from the holder and dispose of it according to your laboratory protocol.
How to Maintain the BioSystems BTS 310 Analyzer
To maintain the BioSystems BTS 310 Analyzer, you need to follow these steps:
Clean the exterior of the analyzer with a soft cloth dampened with water or mild detergent. Do not use organic solvents or abrasive materials.
Clean the cuvette holder with a cotton swab moistened with distilled water or ethanol. Do not touch the optical components with your fingers.
Replace the printer paper and ink ribbon when they run out. Follow the instructions on the printer cover.
Calibrate the analyzer regularly by using standard solutions. Follow the instructions on the user manual[^1^].
Check the battery level of the analyzer by pressing BATT on the keyboard. If it is low, replace it with a new one of the same type.
Troubleshooting Tips for the BioSystems BTS 310 Analyzer
If you encounter any problems with the BioSystems BTS 310 Analyzer, you can try these tips:
Check if there is any error message on the screen. If there is, follow the instructions on how to fix it.
Check if there is any power failure or voltage fluctuation. If there is, turn off the analyzer and wait for a few minutes before turning it on again.
Check if there is any blockage or leakage in the cuvette holder or reagent tubes. If there is, clean them or replace them as needed.
Check if there is any contamination or degradation of the reagents or samples. If there is, discard them and use fresh ones.
Check if there is any interference or noise in the measurement. If there is, repeat the analysis with a blank cuvette or a different sample.
Contact Information for Technical Support
If you need further assistance with your BioSystems BTS 310 Analyzer, you can contact technical support by using these methods:
Phone: +34 93 223 09 18
Fax: +34 93 223 12 08
Website: www.biosystems.com 061ffe29dd